TEMPer8 is a temperature acquisition device connecting with the computer via USB. After connecting up to Four external probes, cooperating with the computer software or computer TXT documents, it can read the ambient temperature data and records it into the computer. (the master device can be connected with Dx, Tx, HS10 probes at the same time !)
TEMPer8 can be used in computer room, weather, chemical reagent and special environment temperature data acquisition.
2. Function & Feature
· Electricity:< 5mA;
· Support Windows NT,XP, VIST;Win7,Win8
· Memory requirement: 20M
· Data auto save to be txt Or csv form, can be open by excel;
· Send mails function
· Graph transfer function
(1) TXT Special button:
As long as you long pressing the TXT button for 1 seconds, you can read the temperature no matter in excel, text or word form without running software. If to stop the reading, press the TXT button again. Easy operation! Solve your installed software and cross-platform used troubles!
(2) Caps lock and Num Lock:
You don’t want to install software, OK! You can use the Caps lock and Num lock function.
As long as you long press the caps lock or Num lock for 3 seconds, you can read the temperature no matter in excel, text or word form without running software. Then you can create curve. (Note: when you use this function, please don't do other action. otherwise, your computer will appear abnormal phenomena. )
(3) PCsensor USB series products, can be used on the mobile phone through OTG interface line.
Q1: What are these products mainly include?
A: TEMPer2 ,TEMPerNTC, TEMPer1F, TEMPer1F_H1, TEMPerHUM, TEMPer1K3, TEMPer1K4, TH1000 ISO, TEMPerF, TEMPe8 series (TEMPer8_D2, TEMPer8_D4 TEMPer8_D8,TEMPer8_H2 TEMPer8_H4, TEMPer8_H8 TEMPer8_T2, TEMPer8_T4, TEMPer8_T8, etc.)
Q2: what are the common OTG interface of mobile phone?
A: millet, samsung, etc.
Q3: how to validate if your mobile phone is in line with the OTG?
A: Use OTG line connecting a mouse or keyboard, see if the mouse is normally used in conformity with the OTG.
3. Data detection
The resolution and precision of the data will be different by different probes. Please refer to the table below
4. Installing instruction
Software installing instruction: please go to www.PCsensor.com to download the process
a) Uninstall the USB driver
b) Install dotnetfx.exe, if your computer system is Vista, may be you don’t need to install this process. Dotnetfx.exe is framework 3.5 of Microsoft, you can download it from the website of Microsoft;
c) Decompress TEMPer8 V1.0.rar to your computer, and running TEMPer8 V1.0 setup.exe.
Plug the thermometer into the USB port of the computer to connect it and running the TEMPer8 V1.0.exe.
5. Operating instruction (Note: Our software only support windows system .)
Run TEMPer8 V1.0 exe, view the real-time data.
Before use, please connect the probe to the computer and then run the software. If you want to change different probes during the measurement, please restart the software. Software interface as shown in the figure below.
Picture 1 Real-time data.
1) DATA LOG
Record the Real-time Temperature Data and Save Them in the Related File Automatically.
Picture 2 Recording
2) CURVES DISPLAY
TAB of the curve, can load temperature curve. Please note that this software can only be used when connected to the computer, so the curve is loaded when the temperature in real time can be read and generate the temperature curve.
Picture 3 Curve
3) MSN FUNCTION
After the MSN User Log in, His Friends can Operate by Related Tips and Obtain the Temperature Data. Your Friends can obtain the Temperature Data Automatically When He Send “1” in According with Tips. The Temperature Data Including: Real-time Temperature, Maximum Value, Minimum Value and Average Value.
Picture 4 MSN
4) Skype FUNCTION
Picture 5 Skype
5) THE SETTING
In functional Settings page, you can set the data storage, alarm temperature, temperature calibration value and temperature unit, language, data path, read interval, alarm value, etc
Picture 6 Setting
6) EMAIL FUNCTION
Fill in the Related Information of Email and Save it. The Requirement of Send Email: Support POP3/SMTP and Other Server Port. Receive Email you can fill in two.
Related Function: 1) when the Alarm is on, it will send the Email to Specified Email Box Automatically if the Temperature Reach the Alarm Limit. 2) You can set the Sending Interval, and then it will send the Email to the Specified Mail Box at Set Intervals. (Suggest) 3) it can Send Email Continuously to the Specified Mail Box. (Don’t suggest) .
Picture 7 Email
1.We suggest you to Reboot Software after Changing “Memory Setting” or “Unit Setting”. It will make sure your Setting cans Take Effect.
2. If the Curves Fluctuate greatly, it maybe you do not Reboot Software in Time after Setting Unit.
6. Packing specification
7. Product photo
TEMPer8 (TXT AURO Reading)
Note: If you want to download the latest software, please click here： http://software.pcsensor.com/pc_en.html
Product Code: TEMPer8_HS10